Now I can see the measured data on the monitor of the HPB. Message 6 of This would suggest me that there is communication through the GPIB. Message 4 of Such error can occur if configuration of GPIB is not correct e. Message 3 of A small update on the setup definition I use:
|Date Added:||25 February 2009|
|File Size:||70.68 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 9 of The default error message for error 6 indicates an abort in the GPIB communication. I will give it a try. I looked on the manual for this error and I cound jp4145b find it.
After this I can see the cursor on the bottom of the I V graph on the monitor of the sliding from left to right when settinp a Sweep of Voltage of SMU1 from lwbview to 1 V, pts. This would suggest me that there is communication through the GPIB.
hpa GPIB through LabView write but no read
I will write Keysight, even though this product I believe to be discontinued, maybe they can offer me some support. This is on labviee B, right? A small update on the setup definition I use: Norbert Norbert CEO: However I get this error and nothing is showing up not on the equipment neither in the VI interface in LabView.
At the same time I get an error on lzbview monitor of D00 Here should be what you are looking for.
Moreover I attach some printscreens where it can be seen how I defined the channel and source. Error 6 I get it only when pressing “Take Measurement” button.
Moreover I cannot see the trace of the sweep I expected a trace of 10 MA constant current and voltage ranging from 0 to 1 V not on the monitor of neither on the graph plot of the LabView VI interface. The VI instrument driver works.
After detailing a bit more the process do you still think there might be a connection drop out between the GPIB interface and HP? I am new to the forum and also new to LabView.
I really appreciate all the feedback I can receive. Message 7 of Message 3 of If i were you, i would contact KeySight formerly known as Agilent formerly known as HP what that message on the device is meaning.
Agilent Technologies / Keysight Technologies hp4145 Analyzer
lavview What exactly is stopping us from doing this? I would like to know if there is any VI available for 2 terminal devices in this case on wafer diodes so that I can control directly the equipment from LabView and be able to read out the measured data directly from the HP via LabView?
The data I want to represent is IV curves. The rest of the channels are not assigned, as now I only try one port measurement, to see if I can take out data from the HP Do you have any up4145b with this?
I have no experience with LAbView so far. I managedto get communication between the equipment HP and LabView. Thanks for the feedback.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Now I can see the measured data on the monitor of the HPB. Message 2 of The only input commands I did were to set up teh channel definition and source setup.